Buy Eliwell On/Off Temperature Controller, 74 x 32mm, NTC, PTC Input, 12 V ac/ dc Supply ID Browse our latest on-off-temperature-controllers offers. Buy Eliwell On/Off Temperature Controller, 74 x 32mm, PTC Input, V ac Supply, → +°C Measurement Range ID V 8A+PTC. Browse our. In order to configure the instrument for this purpose you must access the folder identified by the “Add” label and use the parameters “dEA” and “FAA”. ID LX.
|Country:||Moldova, Republic of|
|Published (Last):||5 June 2008|
|PDF File Size:||10.51 Mb|
|ePub File Size:||1.55 Mb|
|Price:||Free* [*Free Regsitration Required]|
In normal conditions, the labels for the Set point value is found in the menu.
Single stage controller for temperature – ID – Eliwell France
The Setpoint value appears on the display. Set point setting folder; -Pb1: The password is enabled if the value of the PA1 parameter is not 0.
Download from reset Connect the copy card when the instrument is Elisell. The programming parameters are downloaded when the device is switched on.
At the end of the lamp test, the following messages are displayed for about 5 seconds: There are 2 operating modes determined by parameter AtP: The Setpoint can also be viewed. If one or more parameters are modified, it is suggested to the switch off the controller after the modification and then switched back on.
Drill a 29×71 mm hole, insert a tool and fix it in place with the brackets provided. Do not assemble the instrument in excessively humid or dirty locations since it is designed to be used in locations with normal pollution levels. Always make sure that the area next to the cooling openings of the tool is adequately ventilated. Always switch off machine before working on electrical connections. The instrument has screw terminals for connecting electrical cables with a maximum diameter of 2.
The relay contacts are voltage-free. Do not exceed the maximum current allowed.
Eliwell ID970 MC (Milk)
For higher loads, use a suitable contactor. Make sure that the power voltage complies with the device voltage. The sensor has no connection polarity and can be extended using an ordinary bipolar cable note that extending the probe may affect the electromagnetic compatibility EMC of the id90 Probe cables, power supply cables and the TTL serial cable should be kept separate from power cables.
Maximum possible setpoint value. Minimum possible setpoint value.
The two sets are interdependent: HSE maximum set cannot be less than LSE minimum set and vice versa Compressor activation time in eliwsll event of a faulty probe. Compressor off time in the event of a faulty probe. The parameter indicates that a protection is active on the relay actuations of the generic compressor.
Between the request and effective activation of the compressor relay, at least the specified time must elapse. Delay after switching off. The parameter indicates that the protection is active on compressor relay actuations.
At least the indicated time must elapse between switch-off of the compressor relay and the eliweell switch-on. The indicated time must elapse between two subsequent switch-ons of the compressor. Delay time in activating the outputs after switch-on of the instrument or after a power failure. Defrost counting is always active when the machine is on and starts at each power-on. Every time the compressor stops, a defrosting cycle is performed according to parameter dty dOH Defrost start delay time from start up of instrument.
Temperature value understood as distance from the Setpoint or as eilwell absolute value based on Att which if exceeded in an upward direction triggers the activation of the alarm signal.
Temperature value understood as distance from the Setpoint or as an absolute value based on Att which if exceeded id70 a downward direction triggers the activation of the alarm signal. Alarm exclusion time after instrument is switched on leiwell a power failure.
Temperature alarm signal delay time. It is still possible to enter parameter programming and modify the parameters, including the status of this parameter, in order to allow keyboard unlocking. When enabled value other than 0it constitutes the access key for level 1 parameters.
View with decimal point. Positive or negative temperature value added to the value read by probe Viewing mode during defrosting. Programming parameter transfer from instrument to Copy Card.
Programming parameter transfer from Copy Card to instrument. Erasing all parameters in the key. The operation cannot be cancelled. Users must not be able to access parts with dangerous voltage levels under normal operating conditions. The device must be suitably protected from water and dust according to the specific application and only be accessible using special tools except for the front keypad.
It is classified as follows: It should be noted that the relay contacts supplied with the device are functional and therefore exposed to potential elwell. Any protection devices required to comply with product requirements or dictated by common sense due to obvious safety reasons should be installed externally. TTL for connection to Copy Card 3 outputs on relays: This means, for example, that an error introduced by the probe is added to any error that is typical of the instrument.
Although Eliwell has taken all possible measures to guarantee the accuracy of this document, it declines any responsibility for any damage arising out of its use. The same applies to any person or company involved in preparing and writing this document. Eliwell reserves the right to make any changes or improvements without prior notice and at any time.
It’s a community-based project which helps to repair anything.