This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||17 May 2010|
|PDF File Size:||13.24 Mb|
|ePub File Size:||15.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
Printed in the U. Table of Contents Section 1 Specifications Table of Contents Section 5 System Startup Section 1 Specifications Specifications are subject to change without notice. Table 1 Turbidimeter Specifications Nephelometric light scatter at 90 degrees relative to the incident light beam.
The incident light Method of Detection beam is composed of a tungsten filament light source with a color temperature between and K. Section 2 General Information 2. Pay attention to all danger and caution statements.
Failure to do so could result in serious injury to the operator or damage to the equipment. To ensure that the protection provided by this equipment is not impaired, do not use or install this equipment in any manner other than that specified in this manual.
See Replacement Parts and Accessories hah page 55 for ordering information. Installation Electrical Installation High-voltage wiring for the controller is conducted behind the high voltage DANGER The instrument must be installed barrier in the controller enclosure. The barrier must remain in place unless a by qualified technical personnel qualified installation technician is installing wiring for haxh, alarms, or relays.
Seat insulation against connector with no bare wire exposed. Wire gauge less than 18 AWG is not recommended. The controller contains three relays designed for use with high voltage Danger: Relay loads must be resistive. Disconnect and remove the existing wires between the quick connect and terminal block J5, see Figure 21 on page 4. Remove the quick connect fitting and wires and install the nach plug on the opening to maintain the environmental rating.
Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure. Close and secure the cover. Allow at least mm The turbidimeter sensor must be mounted within six feet of the controller unless an extension cable is used. Maximum cable length is 9. Section 4 Operation 4.
The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value. Measurement Xfer Transfer 4. The time is available only in hour military format. In addition, the passcode also regulates function selection for relay options. The hac is factory set to sc the five digits must be followed by a space to remove the trailing asterisk.
Customize the outputs using the table in section 4. To hold the output until released: The data logs store the measurement data at selected intervals. The event log stores a variety of events that occur on the devices such as configuration changes, alarms, and warning conditions. Section 5 System Startup 5.
Push in and turn to secure the connection.
Tug gently to check the connection. After all plumbing and electrical connections have been completed and checked, place the head on the body and supply power to the system. Rinse with deionized water before calibrating. Drain body and clean the body and bubble trap. Measured reading based hsch a gain of 1.
System Startup Two types of verifications are offered.
Hach sc100 1720E Instrument Manual
The calibration history log shows the gain value, the time and date of the calibration, and the initials of the operator cs100 verification. Depends on sample Clean the sensor characteristics. Calibrate Sensor as required by regulatory agency Per agency-dictated schedule.
The hah will depend on the nature and concentration of dissolved and suspended solids in the sample. Restore sample flow to the instrument.
Under normal use, Hach recommends replacing the lamp once a year to maintain peak performance.
Hach-Lange SC 100 User Manual
Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb. Twist the bulb in a counterclockwise direction, pulling out slightly, until it is released from the housing.
Pull the lamp leads and connector through the hole in the lamp housing. Do not touch the new bulb with bare hands. Maintenance Fuse Replacement The controller contains two mains fuses. Failed fuses are an indication that an equipment problem could exist. Problem resolution and fuse replacement should only be performed by qualified service personnel. Refer to Figure 29 and perform the following steps to replace the fuses: Section 7 Troubleshooting 7.
Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions.
HACH SC E INSTRUMENT MANUAL Pdf Download.
Inadequate bubble removal from Increase the signal averaging time to a longer interval. To locate the representative nearest you, send e-mail to intl hach.
Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair. Please contact the Hach Service Center serving your location. In the United States: In the event that a defect is hachh during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Index Numerics 90 Degree Detector Index Sample Line Installation Page of 72 Go. Table s1c00 Contents Add to my manuals Add. Troubleshooting Section 7 Troubleshooting Vis spectrophotometer for hach and hach lange water analysis methods 24 pages. Section 1 Specifications Section 1 Specifications Specifications are subject to change without notice. Theory Of Operation General Information 2.
Controller Dimension Illustrations Installation 3. Mounting The Controller Installation haxh. Page 20 Installation 3. Page 23 Allow at least mm Turbidimeter Installation Information Installation 3. Section 4 Operation Section 4 Operation 4. Controller Display Features Operation 4. Instrument Setup Operation 4. Specifying The Sc010 Language Operation 4. Changing The Sensor Name Operation 4. Setting Up System Security Operation 4.
Page 35 Operation 4. Sensor Setup Menu Operation 4. Page 39 Operation 4. Dry Verification System Startup Two types of verifications are offered. Wet Verification System Startup 5. Replacing The Lamp Assembly Page 50 Maintenance a. Section 7 Troubleshooting Section 7 Troubleshooting 7.
Troubleshooting Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions. Section 11 Limited Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product sv100 refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Index Index Numerics cs100 Degree Detector Page 71 Index Sample Line Installation Page 72 Visit http: Comments to this Manuals Your Name. Enter text from picture: By pressing ‘print’ button you will print only current page.